DIXELL XR20C PDF
View and Download Dixell XR20C installing and operating instructions online. Digital controller with off cycle defrost. XR20C Controller pdf manual download. View and Download Dixell XR20CX installing and operating instructions online. Digital controller with off cycle defrost. XR20CX Thermostat pdf manual. QVWDOOLQJ DQG 2SHUDWLQJ,QVWUXFWLRQV Digital controller with off cycle defrost. SET: To display target set point; in programming mode it selects a.
|Published (Last):||3 February 2011|
|PDF File Size:||15.86 Mb|
|ePub File Size:||2.25 Mb|
|Price:||Free* [*Free Regsitration Required]|
To display target diell point; in programming mode it selects a parameter or confirm. By pressing ‘print’ button you will print only current page. By pressing the 8 key again or by waiting 5s the normal display will be restored. Page of 4 Go. The instrument is fully configurable through special parameters that can be easily.
The instrument must not be. To confirm the operation the “rSt” message starts blinking and the normal temperature. Parameter “IdF” controls the. Push the DEF key for more than 2 seconds and a manual defrost will. Enter text from picture: Each LED function is described in the following table. To change the Set value push the 9 or 8 arrows within 10s.
Page 4 Ptb Map disell Dixell S. To memorise the new set point value push the SET key again or wait 10s. XR20CX panel mounting in a 71x29mm panel cut-out Xe20c Consider the maximum current which can be applied to each relay see Technical Data. Push the SET key for more than 2 seconds to change the Set point value. To print the manual completely, please, download it. Instrument XR20CX shall be mounted on vertical panel, in a 29×71 mm hole, and fixed using the special bracket supplied.
Comments to this Manuals Your Name. Ensure that the wires for probes, loads and the power supply are separated and far. In case of applications in industrial environments, the xr220c of mains filters our mod. Hold press the SET key for more than 3s, while the max.
Defrost is performed through a simple stop of the compressor. Do not expose to water or moisture: The instrument shall xr200c be used for purposes different from those described hereunder.
Dixell XR20C Installing And Operating Instructions (Page 5 of 5)
In case of failure or faulty operation send the instrument back to the distributor or to. DOWN To see the min stored temperature; in programming mode it browses the.
To see the max.
Check the supply voltage is correct before connecting the instrument. Got it, continue to print. Installing and Operating Instructions.
Don’t have an account? The “Lo” message will be displayed followed by the minimum temperature recorded. Page 2 Energy Saving cycle. It provides a relay output to drive the. Fit the probe where it is not accessible by the End User.
By pressing xxr20c 9 key again or by waiting 5s the normal display will be restored. In case of fault in the thermostat probe dicell start and stop of the compressor are timed through.